BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||28 August 2017|
|PDF File Size:||8.49 Mb|
|ePub File Size:||14.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Optimum analysis results can only be obtained with quality tips. Pipetting Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Pipetting c Let the pipetting button slide back slowly. Table of Contents Add to my manuals Add.
Safety Instructions Safety Instructions Please read the following carefully! Checking The Volume Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months. Just send in the instruments to be calibrated, accompanied by an indication of which kind of calibration you wish.
Set nominal volume Set volume to the maximum volume indicated on the instrument see page 32 for procedure.
Enter text from picture: If the pipette operation is clearly inaccurate, or if the instrument must be adjusted for solutions of different densities and viscosities or specially- shaped pipette tips, adjustments can be made using the Easy Calibration Technique. Hold the trwnsferpette vertically, with the sample in the tip for approximately 10 s.
BRAND Transferpette S Operating Manual
Ensure that the transferpettd is securely seated. To print the manual completely, please, download it. Laboratory Equipment Brand dispensette Operating Manual pages. Ask your supplier or manufac- turer for the bfand. Inspect the pipette tip cone for damage. This instrument may sometimes be used with hazardous materials, opera- tions, and equipment.
Pipette tips are disposables items! Eject the pipette tip. Comments to this Manuals Your Name. Don’t show me this message again. Safety Instructions Please read the following carefully! Limitations of Use The instrument is intended for the pipetting of liquids within the following limitations: It is the responsibility of the user of this pipette to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Fitting the tip Use the correct tip according transferpetye the volume range or the color code.
ISO prescribes rinsing the pipette tip once with the sample liquid prior to the actual pipetting process. Adjustment – Easy Calibration Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions. Limitations Of Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity. Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months.
Page 8 Pipetting c Let the pipetting button slide back slowly. Pipette shaft Tip cone Fig.
Autoclave the complete pipette without any further disassembling. Got it, continue to print. Page 9 Trransferpette Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop.
Discharge sample a Place the pipette tip against the wall of the vessel. Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions. Consult the manufacturer for use in temperatures outside of this range. For further information, refer to the accuracy table on pages 37 and Parts will differ slightly depending on nominal volume of instrument. Inspect the piston and O-Ring-seal bfand contamination.
Also See for Transferpette S Operating manual – pages. Page 17 Inspect the pipette tip cone for damage.
Warranty Information Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
Don’t lay the instrument horizontal when the tip is filled. Tranfserpette order for the liquid to reach its end position, leave the tip immersed for a few seconds.
The form can also be downloaded from www. By pressing ‘print’ button you will print only current page. It is beyond the scope of this manual transferpettf address all of the potential safety risks associated with its use in such applications. Page of 25 Go.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Inspect the piston and seal for contamination. Warranty We shall not be liable for transferpete consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
To do this, affix a tip, and aspirate a sample. Don’t have an account? Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. The cycle can, however, be adjusted to individual requirements.